FIU-932: REAR PANEL CONNECTIONS
P1: DC POWER INPUT
This input should only be used with the power supply provided: PS-932-6V-INT.
The 6V regulated power supply for FIU-932 is CE marked and includes four plugs for US/ Japan, EU, UK and Australia.
P2: AUXILLARY FUNCTIONS - SERIAL AND TTL OUTPUTS
This port contains the TTL and serial output signals.
- For the serial output, use cable CB-932-SER-1.
- For use with a parallel port, use cable CB-932-PAR-1.
It is 36 pin MDR type connector. An example mating connector is made by 3M and listed as model number 3M 10136-3000PE.
P3: USB OUTPUTS
For most users this is the main data output port.
It is connected to the host computer using a commonly available USB cable (5 pin mini-USB to USB).
P4: OPTICAL TRIGGER INPUT
This port accepts optical trigger signals such as those produced on some Siemens MR scanners. It uses an Agilent HFBR connector.
Use with cable HFBR-RNS010Z
P5: OPTICAL PORT
This port connects to a Current Designs removable fiber optic bundle or directly to an optical handheld device.
It has 12 optical pins: 4 transmitting and 8 receiving.
P6: TTL TRIGGER INPUT
This connector accepts TTL trigger signals. Triggers are detected on low-to-high (positive-going) edges; there is no minimum pulse width.
Use with cable CB-SMB-BNC-1.